ASSESSMENT OF EFFECTIVE DOSES BASED ON VARIOUS RADON MEASURING TECHNIQUESNataša Smrekar
, 2016, master's thesis
Abstract: In my master's thesis, I have focused on radon gas in 43 buildings used for different purposes (23 schools, 3 kindergartens, 16 offices and a residential house) in which preliminary measurements had shown higher concentrations of radon gas. I carried out measurements of radon and short-lived radon products whilst simultaneously employing different measurement techniques. In all 43 buildings, I measured the instantaneous concentration of radon by using scintillation cells; in 18 buildings, I additionally measured the average concentration of radon by using solid state nuclear track detectors and in 10 buildings, I measured concentration retrospectively with solid state nuclear track detectors. In four selected buildings (a school, a kindergarten, an office and a residential house), I carried out the measurements by using all of the available equipment. I monitored the daily fluctuations of concentration of radon and short-lived radon products by using continuous monitors in 14 buildings. This is how I obtained the factor of radioactive equilibrium between radon and its short-lived products. Based on the results obtained, I calculated the effective doses. As the basis for calculating the doses, I used the instantaneous and average concentrations of radon and the equilibrium factor taken from literature (0.40) or own measurements. I compared the doses and critically evaluated them.
In contrast to the previous research, I researched radon exclusively in areas with increased risk for radon. I studied the influence of the working regime on the concentration of radon in different working environments (i.e. a school, a kindergarten and an office).
Found in: ključnih besedah
Summary of found: ...how I obtained the factor of radioactive equilibrium between radon and its short-lived products. Based...
Keywords: Radon, short-lived radon products, measurement technique, scintillation cell, solid-state nuclear track detector, retrospective detector, equilibrium factor, effective dose, comparison.
Published: 28.09.2016; Views: 4802; Downloads: 255
Fulltext (1,86 MB)