Trace detection of C2H2 in ambient air using continuous wave cavity ring-down spectroscopy combined with sample pre-concentrationOrr-Ewing Andrew
, Damien Martin
, Iain White
, Roberto Grilli
, Ruth Lindley
, Manik Pradhan
, 2008, original scientific article
Abstract: Continuous wave cavity ring-down spectroscopy (cw-CRDS) coupled with sample pre-concentration has been used to measure acetylene (C2H2) mixing ratios in ambient air. Measurements were made in the near-infrared region (λ∼1535.393 nm), using the P(17) rotational line of the (ν1+ν3) vibrational combination band, a region free from interference by overlapping spectral absorption features of other air constituents. The spectrometer is shown to be capable of fast, quantitative and precise C2H2 mixing ratio determinations without the need for gas chromatographic (GC) separation. The current detection limit of the spectrometer following sample pre-concentration is estimated to be 35 parts per trillion by volume (pptv), which is sufficient for direct atmospheric detection of C2H2 at concentrations typical of both urban and rural environments. The CRDS apparatus performance was compared with an instrument using GC separation and flame ionization detection (GC-FID); both techniques were used to analyze air samples collected within and outside the laboratory. These measurements were shown to be in quantitative agreement. The indoor air sample was found to contain C2H2 at a mixing ratio of 3.87±0.22 ppbv (3.90±0.23 ppbv by GC-FID), and the C2H2 fractions in the outside air samples collected on two separate days from urban locations were 1.83±0.20 and 0.69±0.14 ppbv (1.18±0.09 and 0.60±0.04 ppbv by GC-FID). The discrepancy in the first outdoor air sample is attributed to degradation over a 2-month interval between the cw-CRDS and GC-FID analyses.
Found in: ključnih besedah
Summary of found: ...volume (pptv), which is sufficient for direct atmospheric detection of C2H2 at concentrations typical of... ...Enhance Absorption Spectroscopy, Adsorbent Trap, Trace Atmospheric Constituent, CRDS Instrument...
Keywords: Rotational Line, Cavity Enhance Absorption Spectroscopy, Adsorbent Trap, Trace Atmospheric Constituent, CRDS Instrument
Published: 15.07.2019; Views: 2473; Downloads: 0
Fulltext (363,50 KB)